Karl Fischer ROTI®Hydroquant water standard 10.0, 10 x 8 ml ampoules
ADR 3 III
€62.65/Pack Qty. Campaign price!
excl. VAT. | 10 unit(s) per Pack Qty.
Art. No. 5219.2
Now recurring orders conveniently delivered as a subscription!
With the new Carl Roth Replenishment Service you can let products be ordered automatically which you need regularly in your lab!How it works:
Put all products for your subscription in the desired quantity in the basket.
In the shopping cart, select the option "Order shopping cart as subscription" Order as subscription.
Set Starty point and interval for your subscription and submit order!
By the way: Through your account you can customize or delete your subscriptions anytime.
Do you have a question about this product?
- Subtotal: 0.00
|Order No.||Pack Qty.||Pack.||Packaging||Price||Quantity|
|5219.2||10 unit(s)||glass||10 x 8 ml ampoules||€62.65||
Available at short notice
Delivery date currently unknown
There are two established Karl Fischer methods of calculating the water content of a substance: Volumetric analysis and coulometric analysis.
Which method is chosen depends primarily on the water content that the sample is expected to have. Making the right choice is vital for obtaining reliable and reproducible results. Carl ROTH offers compatible reagents for both of these methods.
Karl Fischer volumetric analysis is recommended if the water content of a sample ought to be between 0,1 % and 100 %. Volumetric analysis is performed by measuring the amount of Karl Fischer reagent that needs to be added to a substance until the end point of the titration has been reached. In alcoholic solutions, water reacts 1:1 with iodine in the presence of sulphur dioxide. Methanol and sulphur dioxide form an acid ester that is neutralised by the base (e.g. imidazole, hereafter referred to as "RN" for simplicity).
CH3OH + SO2 + RN → [RNH]SO3CH3
In the titration process, iodine is added which reacts with water to be reduced to colourless iodide:
[RNH]SO3CH3 + H2O + I2 + 2 RN → [RNH]SO4CH3 + 2 [RNH]I
This reaction continues until there is no water left. The end point is determined by a change in colour due to a surplus of iodine and measured using a potentiometer.
Water standards with a defined water content are required to determine the titre for the reagents used (C or T components). Either a solid or a liquid standard can be chosen.
|Water content||10,0 ±0,1 mg H2O/g|